Welcome to our new website! If this is the first time you are logging in on the new site, you will need to reset your password. Please contact us at firstname.lastname@example.org if you need assistance.
Your membership opens the door to free learning resources on demand. Check out the Member Knowledge Center for free webcasts, publications and online courses.
Hear from leaders around the globe as they share insights about their experiences and lessons learned throughout their certification journey.
RAPS recognizes that the current situation in Ukraine impacts our members and customers on many levels. If you are directly impacted by the current situation in the region and are challenged to meet your deadlines or obligations to RAPS, please reach out to email@example.com so that we can defer those challenges. Your health and safety are paramount to us.
Posted 30 September 2013 | By Alexander Gaffney, RAC,
Amongst the long list of reasons for medical device product recalls-manufacturing mishaps, faulty designs, programming errors-few seem as odd as the reason behind a recall initiated on 27 September 2013: The Colorado Rockies exist.
Well, maybe not the Rockies, per se, but the rationale behind Carfusion's Class 1 recall for its Avea Ventilator product is pretty much the same.
In the voluntary recall notice posted on the US Food and Drug Administration's (FDA) website, the company explains that it is initiating the recall because it had identified a "potential risk" when the ventilator is used at high altitudes.
"A proactive complaint review identified an error in the barometric pressure compensation," the firm wrote.
In its recall letter to physicians, Carefusion elaborated further, saying the devices could underreport the tidal volume present in the device when used in conjunction with a flow sensor. The errors can exist at any altitude in excess of 5,000 feet above sea level, the company said. In other words, almost the exact elevation of Denver, Colorado-the "Mile High" city.
The recall was first communicated to its customers on 3 September 2013, and reportedly affects 19,546 units worldwide.
Tags: Recall, medical device